This article through to the understanding of the working principle of the commonly used 20 kinds of liquid level meter
, from the level gauge installation to use, and the analysis of the matters needing attention, to determine the level of possible fault phenomenon and how to deal with, let instrument system understanding of liquid level meter, to meet working conditions to select the level, which provides the basis for make accurate judgment.
Common type level gauge 1, magnetic flap the level 2, 3 float liquid level gauge, steel belt level 4, radar level meter 5, magnetostrictive liquid level meter, rf admittance level 6, 7, a tuning fork level meter 8, glass/glass tube liquid level gauge 9, static pressure type level gauge, pressure, capacitive liquid level meter liquid level transmitter
11 12 13, buoys, intelligent electric buoy level gauge liquid level meter, float transfeeder at 15, 14 electric contact liquid level meter 16, magnetic susceptibility, double color electronic liquidometers 17, outside the level 18, 19 static pressure type level gauge, ultrasonic level gauge, differential pressure type level gauge (20
Double flanged liquid level gauge)
Commonly used the principle of liquid level meter, magnetic flap 1 magnetic level gauge flap level gauge: also called inspcetion level gauge, magnetic level gauge double column.
Principle: the principle of communicating vessels, according to the principle of buoyancy and magnetic coupling developed, when the measured liquid level fluctuation of the container, float in the permanent magnets by magnetic coupling to a magnetic indication panel, double column turned red and white column flip 180 & deg;
, when the liquid level rising column from white to red, when the liquid level drops from red to white, double column panels on the red and white border for container liquid level of the actual height, so as to realize liquid level display.
2, the structure of the ball float liquid level float liquid level gauge mainly of design and production on the basis of the principle of buoyancy and static magnetic field.
With the floating ball of magnets (
In the position of the measured medium affected by buoyancy effect: the change of liquid level lead to the change of the magnetic float position.
Float in the magnet and sensors (
The magnetic reed switch)
Function, make the string into the circuit components (
Such as fixed value resistor)
The number of changes, and then cause changes in the amount of electricity meter circuit system.
Is the change in the position of magnetic float cause the change of the electrical signal.
By measuring the quantity of electricity to reflect the container level.
3, steel belt level meter which is fabricated using the mechanical balance principle design.
When the liquid level changes, the original mechanical equilibrium under the float by the buoyancy of the disturbance, will move through the steel strip of reaching a new equilibrium.
Liquid level detection device (
According to the level of leading steel belt move, the displacement of the transmission system working through the movement of the steel belt drive pin rotation, which ACTS on the counter to display the level of the situation.
4, radar level gauge radar level meter is the measuring instrument based on the principle of time travel, radar wave running at the speed of light, running time can be transformed into material level signal by electronic parts.
Probe a high-frequency pulse and probe spread along the cable type, when the pulse in the material surface reflected by the receiver inside the instrument, and the distance signal is transformed into material level signal.
5, magnetostrictive liquid level meter of magnetostrictive liquid level sensor, the sensor circuit part will inspire a pulse current on waveguide wire, wire when the current wire along the waveguide propagation in the waveguide of impulse current magnetic field generated around.
Outside the magnetostrictive liquid level sensor rod is equipped with a float, the float along the rod can move up and down along with the change of liquid level.
A group of permanent magnetic ring inside the float.
When the impulse current magnetic field when they met with float copper produced by the magnetic field, the magnetic field around the float is changed so as to make the wire made of magnetostrictive materials waveguide in float position to produce a torsion wave pulse, the pulse at a fixed speed along the waveguide wire back and detected by the detection institution.
By measuring the pulse electric current and time lag of the torsion wave can accurately determine the position of the float is located, namely on the surface of the liquid position.
6, rf admittance level rf admittance level meter is composed of sensor and control meter, the sensor can be used rod type, or coaxial cable type agent installed on warehouse top.
The pulse sensor card can convert level changes to the pulse signal to control instrument, control instrument after processing into a display of quantities, so as to realize the continuous measurement of material level.
7, a tuning fork level meter level controller tuning fork type of works by installed in the fork of a piezoelectric crystal that tuning fork under certain resonance frequency vibration.
When tuning fork and contact measured medium, the frequency and the amplitude of a tuning fork will change, these changes to be inspected by intelligent circuit, processing and converted to a switch signal.
Guangdong Kaidi Energy Technology Co., Ltd. has various branches in local businesses, servicing customers and helping to pull in traffic to those businesses.
All of the long-term strategies and short-term actions of KAIDI will be molded by a set of core values that are shared by each and every associate.
level gauge are raising the stakes of social marketing, but they also ease the sales process by providing ways for customized level indicator
to effectively interact with customers.
Lucky to know that you are not alone in the face of customized level indicator issue. Let Guangdong Kaidi Energy Technology Co., Ltd. be your selected level gauge expert in providing first class to help you out.