19…60VDC 19…250VAC 50/60Hz
In the medium 2 seconds, empty: 3 seconds
Powder and particle =10mm，density>0.1g/cm³
Liquid, viscosity =10000mm2/s, density >0.7g/cm³
Relay, AC250V/4A, DC60V/4A
DC<3W, AC <15W
● Insert the fork into the medium to be measured about 10mm depth, Press the button “SET” until red and greed LEDS blink alternatively for at least 2 times, then release the button “SET”, now red LED blinks, then press the button “SET”, red and greed LEDS blinks alternatively then both on.
● Remove the fork from the medium, green LED on, and red LED off (NC is closed with COM, NO and COM are opened), now insert the fork into the medium again, Both LEDs are ON, (NO is closed with COM, NC and COM are opened). Now setting are finished.
● Default density is set for water: 1g/cm³, If the target medium density ≥ 1g/cm³, Self-Checking is not necessary. Self-Checking is necessary when the target medium density < 1g/cm³.
● Default Sensitivity is the highest, for material that have irregular surface and fluctuate, please set lower Sensitivity.
● Avoid to install in the position that vibrate drastically.
Please Send Us your Application details
Measuring range , medium, Installation method (flange, thread), Pressure range, Remote transmitter, Level alarm switch
Our technical team will check your specifications and make a best solution for you.
More Model available for all appication, different termperture and pressure!
We are here to help you! If you close the chatbox, you will automatically receive a response from us via email. Please be sure to leave your contact details so that we can better assist