The installation of the electromagnetic flowmeter
is a systemic work, * first asks the user to the installation position of this enterprise and its conditions have a comprehensive understanding, and then according to the manufacturer's instructions or * * technical guidance personnel help to correct installation.
This article is for connection and large diameter electromagnetic flowmeter cleaning pipe sensor in the underground sewage pipe installation of the related problems in the process of introduce one to you.
More details please call our sales and service personnel, will give you the guidance of the * *.
Figure 6 to 18 for cleaning of pipe connection on the other hand, for small diameter electromagnetic flowmeter sensor, due to its light weight, can be installed directly in the path of the metal tube.
And larger diameter of the sensor, due to the large volume, its own weight, should fall to the ground installation, or use bearing support.
In supporting bracket, also must consider the fixed support and maintenance personnel activity space.
Five, the large diameter sensor installation due to the large diameter pipe is buried in most situation, so big caliber sensor before installation should be concrete pit.
As shown in figure 6 to 19 is in the installation of water supply and drainage system in the common form.
Should have enough activity space, concrete pit wall of steel pipe and cable laying on the cover to prevent the rain, a drain pipe to avoid water accumulation in the pit and the sensor inundation.
In order to tear open outfit is convenient, the sensor should be put on shoes, and in the downstream side is equipped with adjustable telescopic tube.
Figure 6-20 is the electromagnetic flowmeter in underground concrete pipe installation instance in the sewage system.
Figure 6-19 large caliber sensor installation diagram in figure 6-20 sensors in underground sewage pipe installation 1 - feed line;
2 - the overflow;
3 - the water convex platform;
4 - cleaning hole;
5 - sensor;
6 - removable parts;
7 - wall seal;
8 - outlet pipe;
9 - discharge valve (
Instrument science and technology in this paper by embellish release, if you encounter any problems in the process of using the instrument, can call our domestic sales service hotline: 0517 -
86917118, we have * * personnel unriddling for you)
Many of us have heard about level gauge and seen some of these units in operation in customized level indicator, customized level indicator and customized level indicator spaces.
Guangdong Kaidi Energy Technology Co., Ltd.’s purpose is to create superior value for our customers, employees, communities and investors through the production, conversion, delivery and sale of energy and energy services.
Another way to maintain the professional yet engaging innovative technology in level gauge is by embedding new skills directly on manufacturing.
Overall, level gauge may be a great way for manufacturers to expand their use of technology, but the price could present a significant hurdle for some businesses.