Q: old pipeline, pipe wall scaling is serious, you can't get a signal or when the ultrasonic flowmeter
measurement signal is too weak, how to solve?
A: check whether there is fluid in the pipeline, and for the full state;
Should choose the Z method ultrasonic sensor installation (
If the pipe is too close to the wall, can have Angle of ultrasonic sensors are installed on pipe diameter, rather than the horizontal installed on pipe diameter)
Dense part carefully selected pipes and fully polished bright, apply sufficient coupling agent installed ultrasonic sensor;
Respectively carefully at every ultrasonic sensor is installed near the point of moving slowly, look for the large signal points, prevent because of pipe wall scaling or because local deformation is expected to lead to the ultrasonic beam reflected area and miss can receives the strong point of installation;
For wall scaling serious metal pipe can use hit to make scale parts fall off or crack (
Note: this method can sometimes because of scale and gaps between the lining and the disadvantage and the ultrasonic propagation)
Q: pipe is very new, material is very good, also in line with the ultrasonic flowmeter installation conditions, but did not receive the signal?
A: check whether the ultrasonic flowmeter pipe parameters is set up correctly, the installation method is correct, wiring whether contact in good condition, whether the coupling agent with enough, whether the pipeline for the full package, whether installed in accordance with the instruction of total graphic distance ultrasonic sensors, ultrasonic sensor installation direction is wrong.
Q: why is the pipe flow, and ultrasonic flowmeter host also shows & other;
State, and the machine according to the instantaneous flow rate is zero?
A: check whether in cases with fluid used & other
Static zero setting & throughout;
And, if so, use M43 menu, restore the original factory Settings zero.
Q: why is ultrasonic flowmeter output current loop current value is not normal?
A: check whether ultrasonic flowmeter M55 menu, set up the required output current;
Check the ultrasonic flowmeter M56, M57 menus of the current Settings to the upper and lower limit is appropriate;
Aligning current loop, and use the M49 validation.
Guangdong Kaidi Energy Technology Co., Ltd. has various branches in local businesses, servicing customers and helping to pull in traffic to those businesses.
We are an experienced supplier of and have gained good reputaion among global customers. With a wide range of in offer, we can customize according to your requirement. Send us your enquiry at Kaidi level indicator.
According to the latest social survey, more than 50 percent of consumers (across all age demographics) follow a brand before purchasing a product. Therefore, KAIDI's content can make or break a customer's decision to conduct business with you.
level gauge developed from Guangdong Kaidi Energy Technology Co., Ltd.’s unique skills in high technology has helped to produce level gaugecustomized level indicator.
The more people who do a certain thing, the more likely others are to do it as well. When KAIDI can demonstrate their popularity or satisfaction across a wide customer base, other consumers are more likely to buy in as well.