1, level gauge
shall be installed on the top of the tank, to avoid equipment inside the moving parts of a collision;
When the equipment within the medium volatility against the wave guide rod (
Through hole type protection tube fixed.
2, level gauge on the pipe outside the equipment installation, shall meet the following requirements: a)
Wave guide rod (
Should include the length of the upper and lower measuring dead zone, port department should be at least 50 mm below the lower connecting pipe joint center;
Using double pole type probe guided wave radar level gauge, pipe diameter should not less than 80 mm;
Using single lever probe guided wave radar level gauge, pipe diameter of not less than 50 mm.
3, use cable probe type guided wave radar level meter big liquid level measurement, should be to straighten cable probe fixed at the bottom of the device, liquid level of volatile situation should add protection tube fixed through hole type.
4, high temperature measured medium, should separate the transmitter installation.
5, the installation of guided wave radar and capacitive liquid level gauge should also meet the requirements of the factory.
Guangdong Kaidi Energy Technology Co., Ltd. who is highly knowledgeable about manufacturing as well as selling and confident in our ability to create finest products as level gauge customized level indicator.
Guangdong Kaidi Energy Technology Co., Ltd. humbly request you to try this item in your centers and we assure you that you would be in a great pleasure with the results.
Guangdong Kaidi Energy Technology Co., Ltd. offers not only the high-quality product but also the finest service, gives the customer with an expressive using experience.
Obviously, financial return is important in manufacturing level gauge, but I think that's not enough. I think many customers want to support something they really believe in.