Friend, engaged in chemical and other industries level measurement are touched magnetic flap liquid level gauge
Magnetic level gauge turning on a large number of towers, tank level measurement occasions there are a large number of applications.
Install magnetic flap after the production of liquid level gauge should be level of the place should be to show the red column, no liquid column should display is white, this is the normal work of magnetic level gauge turning status, because the magnetic level gauge is turning of work on the basis of the principle of buoyancy and magnetic coupling liquid level measuring instrument.
But occasionally instrument in container level normal fluctuation position changes, magnetic level gauge panel double column (turning
But remains the same, this is what reason is caused?
For such failure analysis has the following methods: 1, to produce this kind of phenomenon, * need to verify the real value of liquid level specific, first confirm really have the liquid level in the tank.
Opens the diverter valve or check whether really have the liquid level inside the tank, it is only really have the level of our inspection.
2, check the line of the cut-off valve is open.
If confirm with liquid level, then carefully check float, take a look at the float is blocked by some reasons or caton, in after carefully check the float, and do to float right after processing, can open the switch valve, to put the magnetic float level gauge turning into measuring tube inner cavity.
3, according to the principle of magnetic level gauge turning design because it is the level where, float because of the in what position.
Since the critical point of the liquid level and the atmosphere is red column, so for the said float does not get stuck, it's right, the air below the critical point directly to the frame with magnet using magnetic coupling to turn into a red double column.
If again putting-in-service proactively level gauge float to the liquid level drops to zero.
The magnetic rotor, usually be found after can look at the monitor whether have damage, if be in foreign body.
If not, then that is a float.
With the iron on the wall, and confirm the position of the float, confirm the float have no card, and then the lower part of the inspcetion painted red column, column above to white, if customers don't trust, can open the drain valve to discharge, emission and putting-in-service proactively again.
Instrument science and technology in this paper by embellish release, if you encounter any problems in the process of using the instrument, can call our domestic sales service hotline: 0517 -
86917118, we have * * personnel unriddling for you)
Collectively, the effect of level gauge on industrial society has been to eliminate customized level indicator and drastically reduce the time long associated with customized level indicator.
Guangdong Kaidi Energy Technology Co., Ltd. is committed to attracting, developing, and keeping a diverse work force that reflects the nature of our global business.
Do you want to find a provider to get your level gauge problem settled? If so, we suggest that you give a shot to Guangdong Kaidi Energy Technology Co., Ltd.. Visit Kaidi level indicator to learn more and contact us.
Offering a loyalty program not only makes customers feel valued, but it allows Guangdong Kaidi Energy Technology Co., Ltd. to easily collect important information about customers.
Knowing what promotions are popular and get the most activity as level gauge from current and potential customers can play a role in your overall strategy.