For instrument accuracy requirement is to ensure that the premise of enterprise production precision control, for the guarantee of accuracy is not only with the characteristics of the product itself, related technology, sort, and late and instrumentation installation and maintenance methods are inseparable.
In general, the installation of instrumentation products must be based on working principle and structure characteristics of each instrument product, given the corresponding environmental suitability and install the operating skill, instrument installation point of view of the product itself.
Instrument installation process of cuhk error is not related to the characteristics of the instrument is known and installation requirements, assume how how convenient installation, this will not only for instrument use effect and the adverse impact on the service life, will directly influence the outcome of the production.
In this paper, by way of example of metal tube float flowmeter
, metal tube flow meter in the measurement accuracy is guaranteed in the process of use, must be mastered in the installation of four main point: 1, install the magnetic filter and valve, which can be made when using flow meter to measure, can smoothly will exist in the medium with a magnetic material filter, medium if solid impurities, can also be filtered through a filter.
2, metal tube flow meter should be installed on the process pipeline, and beside it must install bypass pipe and bypass valve, to do this also for the detection of maintenance time, better replace flowmeter and cleaning pipe, the metal flow meter daily inspection work more convenient.
3, try to avoid install the instrument in direct sunlight place, it will cause harm to the liquid crystal display instrument, makes the service life of the LCD greatly reduced.
4, install valve in the instrument at the exit, such doing is to be able to when doing the gas flow measurement, not export direct emissions from flowmeter, if direct emissions at the exit, it is easy to cause air pressure is reduced, resulting in data loss.
Research and development of the third generation of metal tube float flowmeter, on the basis of original products formed the following five unique selling point: one time forming inner and outer taper pipe one-piece;
Second, the low pressure loss design;
3 it is to float can be difficult to jam, magnetic components around is not affected by earth's magnetic field;
Four is solid, no dead Angle measuring tube,;
Five is modular, intelligent, card type indicator design, new ESK transmitter of synchronization with the domestic market.
Instrument science and technology in this paper by embellish release, if you encounter any problems in the process of using the instrument, can call our domestic sales service hotline: 0517 -
86917118, we have * * personnel unriddling for you)
Guangdong Kaidi Energy Technology Co., Ltd. highlighted the need to foster a human openness to technological innovation.
Guangdong Kaidi Energy Technology Co., Ltd. will become the destination store for customers, offering the convenience of multiple brands and channels, and providing a personal high touch shopping experience that helps create lifelong customer relationships.
There have been conclusive evidence on 's role in customized level indicator
and customized level indicator.
Guangdong Kaidi Energy Technology Co., Ltd. has never compromised on the quality and the services provided to the customer.
customized level indicator level gauge will help keep your customized level indicator in a customized level indicator state.