Magnetic flap liquid level gauge
, as a kind of intuitive, simple structure, convenient maintenance of liquid level measuring instrument, a large number of applications in tank farm, for its common failure, our technicians summarized the eight methods are very useful, according to these methods, we basically can be found and find the cause of the failure, the eight methods listed below: 1, the hitting hand pressure often encounter the phenomenon of instrument operation and downs, most is the phenomenon caused by poor contact or virtual welding.
In this case with hand pressure can be used.
The so-called & other;
Percussion & throughout;
Is on the part of the fault may be produced by small rubber ð ¨ + head or other tap tap the magnetic level gauge board or turning parts, see if it will cause error or downtime
The so-called & other;
Hand pressure & throughout;
Is when failure occurs, turn off the power to connect components and plug and seat back pressed firmly with the hand, it will try to eliminate the fault.
Tapping the chassis is normal, if found again knock again not normal, good to all joint heavy stuck to try again.
2, observational method using visual, smell and touch.
To some, the damaged element will change color, blister or charred spots;
The smell of burning device will produce some special;
Short circuit of the chip will be hot;
Magnetic flap can be observed with the naked eye level gauge virtual soldering or desoldering.
3, the so-called exclusions method is through some board inside the machine, pull device to determine the cause of the problem.
When removing a card or device instrument returned to normal, after failure there.
4, replacing method requires two same magnetic level gauge or turning have enough spare parts.
Will be a good spare parts with the same components of fault on board to replace, the fault is eliminated.
5, comparative law requires two same instrument, and there is a normal operation.
Using this method also has the necessary equipment, for example, multimeter, oscilloscope, etc.
According to the comparison of the nature of the points, comparison, waveform comparison, comparison of static impedance, voltage output comparison, comparison of current and so on.
The specific method is: let the defective magnetic level gauge and turning normal running in the same situation, and then some detection point of signal be comparative test of two groups of signal, and if there is different, you can conclude that fault out here.
This approach requires maintenance personnel have considerable knowledge and skills.
Sometimes 6, cooling method, the instrument work a long time, or working environment at high temperature in summer can appear fault, check normal shutdown, stop for a period of time it is normal again, fail again in a moment.
This phenomenon is due to individual IC or component performance is poor, temperature characteristic parameters are not up to standard requirements.
In order to find out the cause of the problem, can use litre cooling method.
So-called cooling is when failure occurs, cotton fiber to anhydrous alcohol in wiping may be out of order, make its cool, observe whether fault is eliminated.
So-called warming is artificially environment temperature, such as soldering iron nearer questionable parts (
Attention must not will increase temperature too high, so that damage the normal device)
Whether tested failure.
7, ride a shoulder on shoulder also called parallel method.
Put a piece of good IC chip in to check the chip, or the good components (
Resistance capacitance, diode, triode, etc. )
With components in parallel to check, maintain good contact, if internal open circuit fault from the device or poor contact, that is the way that can be ruled out.
8, capacitance bypass method when a circuit produces quite strange phenomena, such as display confusion, can use the capacitance bypass method about the fault circuit part.
The capacitance jumper in the power of the IC and end;
Jumper in the base of transistor circuit input or collector output, to observe the effects of the failure phenomenon.
If capacitance bypass invalid input and the output of the bypass it end failure phenomenon disappear, then determine the fault occurs in the primary circuit.
In this paper by embellish instrument technology co. , LTD. Arrange release.
The manufacture marketing game of level gauge
is changing with each innovation, and businesses of all products need to be ready to pounce.
Providing highly qualified level gauge products and services, Guangdong Kaidi Energy Technology Co., Ltd. is committed to helping clients make lasting improvements to their performance and realize their most important goals. Over the past decades, we’ve built a firm uniquely equipped to this task. Go to Kaidi level indicator
for more info.
KAIDI is designed to enhance your savings in terms of cost, energy and efforts.If you are interested in our level gauge customized level indicator products, please contact us
If we avoid overselling level gauge customized level indicator and capabilities, you gain credibility and earn trust. We did not want to jeopardize our ingrained preference for humility.